GRASEBY 3100 SYRINGE DRIVER DETAILS:
|File Size:||16.0 MB|
|Supported systems:||Windows 10, 8.1, 8, 7, 2008, Vista, 2003, XP, Other|
|Price:||Free* (*Free Registration Required)|
GRASEBY 3100 SYRINGE DRIVER
Graseby Automatic Syringe Driver Pump Infusion eBay
Whenever the case is taken apart the Syringe Size Sensors tests detailed on page must be carried out. Repair proceduresSmiths Medical International Ltd.
|hp photosmart c3150 printer||Related titles|
|ati 264vt3||Gulmohar Health Care|
Main board renewal1. Open the graseby 3100 syringe see page Remove and retain the six pan head self tapping screws that fix theMain board to the front casing.
User Manual for Graseby 3100 Syringe Pump
Displace and carefully turn graseby 3100 syringe faulty board over. Disconnect the six connectorsfrom the underside of the board; from the left, plug PL7, 4, 2, 1, 6 and 3 Figure 7. Remove the faulty Main board.
Fit a new board graseby 3100 syringe reversing steps 2 and 3 above, and close the casing seepage Regulator boardrenewal1. Make a note of how the eight spade connectors are mounted on the Regulatorboard and then slide and lift the connectors off.
GRASEBY SYRINGE PUMP (USED) - Diac Medical
Prise open the two nylon retaining clips and release the four wires. Remove and retain the two screws that hold the transformer in place.
The lefthand screw is 25 mm long and the right hand screw is 8 mm long. Lift out thetransformer. Remove and retain the two screws that hold the Regulator graseby 3100 syringe in place andthen remove the faulty board. Fit a new board by reversing steps 2 to 6 detailed above, and then close thecasing see page Repair proceduresOpto sensors boardrenewal1.
GRASEBY 3100 SYRINGE FLUID INFUSION PUMP DRIVER
Remove graseby 3100 syringe retain the six pan head self tapping screws that are used to fix theMain board to the front casing and turn the board over. Carefully lift up the retaining clip on PL7 Main board; Figure 7.
Remove and retain the screw and washer that graseby 3100 syringe the ribbon cable clip of PL7to the chassis. Remove and retain the two fixing screws and washers from the front of theOpto board.
Ease out the graseby 3100 syringe flag from the sensors assembly and remove the faultyOpto sensors board. Fit a new Opto board by reversing steps 2 to 6 detailed above, and then closethe casing see page AC transformerrenewal1. Make a note of how the eight transformer spade tags are connected to theRegulator board and then slide and lift the tags off. Prise open the two nylon retaining clips and release the four transformer wires. Remove and retain the two screws and the spacers that hold the transformer inplace. The left hand screw is 25 mm long and the right hand screw is 8 mmlong. Lift out the faulty transformer.
Fit a new transformer by reversing the steps 2 to 4 detailed above and thenclose the casing see page Report details. If a syringe is fitted the associated syrin- ge size LED will remain on. Insert the following test rods consecutively in the syringe cradle:.Thank you for choosing a Graseby Graseby 3100 syringe Pump.
This Instruction. Manual contains important graseby 3100 syringe on operating and looking after your pump. The Graseby Syringe Pump is an operator programmed micro-computer controlled syringe pump to dispense liquids It caters for 20, 30, 50 and 60ml.